Dimethyl Sulfate 0.005/c, 0.005 - 0.05 PPM, (6718701)
Dimethyl Sulfate 0.005/c, 0.005 - 0.05 PPM, Part # 6718701, 9 Tests
Standard Measuring Range: 0.005 to 0.05 ppm Discolouration compared to colour standard.
Number of Strokes n: 200
Time for Measurement: approx. 50 min
Standard Deviation: ± 30 %
Color Change: white blue
Ambient Operating Conditions
Temperature: 15 to 30 °C
Absolute Humidity: 3 to 15 mg H2O / L
Dimethyl sulphate + 4-(4-Nitrobenzyl)-pyridine colorless
colorless alkylation product blue reaction product
Phosgene and chloroformates cause a yellow discoloration of the indicating layer, and it is impossible to perform a dimethyl sulphate measurement. Alcohols, ketones, aromatics and petroleum hydrocarbons in the TLV range do not affect the indication.
After carrying out the required 200 pump strokes the reagent ampoule must be broken, the liquid transferred onto the indicating layer and carefully drawn through it using the pump. Wait five minutes before evaluating the indication. The tube must not be exposed to direct sunlight during the 5 minutes waiting period.