Hydrogen Sulfide 0.2/b, 0.2 - 6 PPM, (8101991)
Hydrogen Sulfide 0.2/b, 0.2 - 6 PPM, Part # 8101991, 10 Tests
Standard Measuring Range: 0.2 to 6 ppm
Number of Strokes n: 1
Time for Measurement: approx. 55 s
Standard Deviation: ± 15 to 20 %
Color Change: yellow pink
Ambient Operating Conditions
Temperature: 15 to 40 °C
In case of temperature between 0 °C and 10 °C, the reading has to be multiplied by 1.5; standard deviation: ± 30%.
Absolute Humidity: max. 20 mg H2O / L
H2S + HgCl2 HgS + 2 HCl HCL + pH indicator pink reaction product
Up to 1000 ppm, sulfur dioxide has no influence on the reading. Within the range of their TLV, mercaptanes, arsine, phosphine and nitrogen dioxide are also indicated, however, with differing sensitivity. Within its TLV, hydrogen cyanide changes the color of the entire indicating layer to a light orange. The reading of hydrogen sulfide is not affected.